Texas Instruments TLC Series Digital to Analog Converters – DAC are available at Mouser Electronics. Mouser offers inventory, pricing, & datasheets for . D Easily Interfaced to Microprocessors. D On-Chip Data Latches. D Monotonic Over the Entire A/D Conversion. Range. D Segmented High-Order Bits Ensure. In order to try out the new I/O expansion bus for the MC3 I built a few quick I/O boards. The first one is a simple 8-bit DAC. I had an TLC laying around.
|Published (Last):||5 April 2015|
|PDF File Size:||17.9 Mb|
|ePub File Size:||15.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
I was lucky that the arrangement of the DAC data pins matched my bus connector. After the buffer amp I am getting 0. No track and trace will be send to the client. This thread has been locked. Ask a new question Ask a new question Cancel. I did some more testing and got interesting results.
Verifying consistancy of tlf7524 MD array under Debian.
In reply to Jaakko Nevala:. The measurement error is documented on a calibration certificate, which serves as proof.
tlc724 Ask a related question What is a related question? Last but not least both specs call the resistance between Vref 15 to ground to be 5k to 20K.
The comment should definitely be in the datasheet. If one were to use a fixed reference tlf7524 then a gain stage, the output may not swing all the way to Vcc, or if Vcc were low, the op amp might saturate at a digital code of well below Mentions Tags More Ltc7524. I have always assumed the AD and the TLC to be interchangeable but from your tests it looks like that is not the case.
A simple DAC using the TLC «
YM playback on the YMZ Tl7c524 The range of this service include, but not limited to: Enclosure for the MC3. Share this product by email Print this page Report an error Report an error Thank you for helping us with our website! I tlx7524 the Tlc and AD have basically the same design. Eugenio Nejia Is the parallel interface necessary for speed? The Korg Monotron Duo. Software defined radio with rtl-sdr. A quite tidy backside. This is what the MK1 output looks like tlf7524 a scope running a simple loop to generate a sawtooth.
In collaboration with Intertek, Distrelec is offering a calibration service for new electronic test and measurement instruments. An identification of the calibration method eg. Thank you for sharing. The TSC spec does not say so. I have already several of these available and time is of the essence. I am using 5V buffered reference and 10V buffered supply voltage.